During a customer training course it was determined that when both of these meters are in DEMO mode - the phasor diagram . Please see atatched pictures. Does anyone know why? What is the fix?
Your PM8000 is in DEMO mode. The Phasor viewer visible from the front panel is a fixed value with slight randomization applied to it (i.e. V2 angle with be approx. 120 degrees +/- 1) while ION Setup is reading the actual value being computed by the Harmonics Measurement modules (same applies for all harmonics displays on the front panel while the device is in DEMO mode).
Discuss challenges in energy and automation with 30,000+ experts and peers.
Find answers in 10,000+ support articles to help solve your product and business challenges.
Find peer based solutions to your questions. Provide answers for fellow community members!